• A stand-alone, high-speed whole rack tube capper/decapper that can automatically remove and then re-cap, or dispose of, septum caps from a full 96 well format tube rack.
• Compatible with 0.5ml, 0.75ml, 1.0ml and 1.4ml septum sealed storage tubes.
The XDC-96 is a stand alone, high-speed whole rack tube capper/decapper that can automatically remove, then recap or dispose of, septum
caps from a full rack of 96 tubes.
Eliminating the need to manually remove the caps dramatically improves productivity and, as septum piercing is no longer required, the the user is able to make use of a wider range of liquid handling systems.
The XDC-96 will also dispose of the caps or recap the same caps back into the same tubes. Manually recapping tubes after sampling is no longer
Cycle time (decapping and recapping) is 15-30 seconds, depending on tube type.
Compatible with most 96 format TPE septum sealed caps (0.5ml, 0.75ml, 1.0ml and 1.4ml).
For more details on integration and the Purge Module see XDC-96PRO.
Ideal with FluidX TPE septum caps.
Type of tube to be used with the XDC-96 capper/decapper must be specified at the time of ordering.
The XDC-96 can only be used with natural coloured TPE caps.
Width 215mm x Height 350mm x Depth 680mm (with shuttle), 480mm (without)
AC 115 V /230V +/- 10%. Power less than 55 W.
Membrane panel for manual operation. Windows compatible maintenance and calibration software package provided.
FluidX 96-format tubes with septum caps (1.4ml, 0.5ml).
Micronic microtubes with capcluster (1.4ml, 0.65ml).
Matrix TrakMates with SepraSeal caps (1.4ml, 0.75ml, 0.5ml).
Abgene Cypher with caps (1.2ml, 0.65ml).
Please contact FluidX for information on additional tube types.
Cat No 46-5001
XDC-96. Automatic 96 well format tube rack capper / decapper compatible with FLUIDX, Matrix / Micronic tubes and Individual caps including 1.4ml, 0.75ml, 0.65ml**, 0.5ml tube variants.
For Benchtop use only, use XDC-96Pro for integration or RS232 control.
Please specify tube range when ordering e.g. FluidX or Abgene or Matrix or Micronic.
(**Some restrictions apply to smaller volume tubes)